|Published (Last):||26 November 2009|
|PDF File Size:||2.70 Mb|
|ePub File Size:||1.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
The calibration message containing the factory calibration date can be retrieved by using the "CALibration:STRing" command.
The flashing digit indicates the digit of the output parameter displayed that can be changed by turning the front panel knob. Use the resolution key to move the flashing digit to the left or right until the flashing stops. The settling time is To disable the front-panel control knob, scroll the flashing digit on the front-panel display to the right or left using the selection keys until the flashing digit disappears. Notice that the control knob and all front-panel keys are n They can be overwritten as many times as desired but there is not an explicit way to delete saved states.
Keysight power supplies are designed to have the lowest noise that is practically possible. For some applications it is desirable to simulate the effect of noise on the output, for example to test system immunity to power line disturbances. The EA, EA, EA and EA have 3 non-volatile recall states for the output parameters, each of which can store the voltage, current limit and over-voltage protection setting.
Both remote If you have a need to install one or more in a rack, order rack mount kit This kits contains a sup When the Output is disabled, press the 'Display Limit' button on the front panel to place the power supply in the limit mode. Then, set the limit values for the output voltage and current. Press the 'Output Off' button again to enable the power su Criteria have been established to determine which products get or do not get a calibration from the factory.
A product with a product number does not automatically qualify for a calibration certificate. Peak-to-peak noise is measured from the maximum positive point on the waveform to the most negative voltage point.
In general, current sinking is a fixed value at some percentage of the full scale current output. However, there is an exception for one series of Keysight power supplies. On the xB series, the maximum negative current tracks the value programme No other programming languages are available. These power supplies have all the features and performance necessary for both benchtop and system ATE applications where throughput is not critical, at a surprisingly low price. When the Output is off, press 'Display Limit' to put the supply into the limit mode and set the limit values for voltage and current.
Then, press 'Output Off' again to set the supply output on. Rackmount Kit - The power supply can be mounted in a standard inch rack cabinet using one of three optional kits. The Keysight ExA offers new versatile binding posts on the front panel and screw type terminals on the rear panel. Where can I find manuals for discontinued and obsolete Hewlett-Packard, Keysight and Keysight test equipment?
FAQ What does the check mark at the rear panel label of the E series bench power supply mean? The main purpose of the yellow label as shown below is to caution customer How can I eliminate the flashing single digit in the front-panel display? What is the settling time for the ExA supplies? The settling time is FAQ ExA and ExA: Disabling the Control Knob To disable the front-panel control knob, scroll the flashing digit on the front-panel display to the right or left using the selection keys until the flashing digit disappears.
How can I clear the content of power supply memory for programmable E36xxA power supplies? How can noise be superimposed on the output of a dc power supply? How much internal memory is available to the user in the ExA and ExA power supplies? Which of the E36xxA power supplies can be controlled from the remote interface? How long is the calibration time interval? Keysight recommends 1 year as the calibration interval for all E series power supplies. How do I rack mount the E36xxA manual and programmable power supplies?
Does the Keysight ExA offer any alternative programming languages? ExA: Bench vs. System Applications These power supplies have all the features and performance necessary for both benchtop and system ATE applications where throughput is not critical, at a surprisingly low price.
ExA: Accessories Rackmount Kit - The power supply can be mounted in a standard inch rack cabinet using one of three optional kits. What types of output connections are allowed?
Keysight / Agilent E3645A 80W DC Power Supply, 35V, 2.2A or 60V, 1.3A